What is the difference between the eSync and eSync 2?
The eSync is a legacy device which has been replaced by the eSync 2. Compared to the eSync, the eSync 2 offers more synchronization ports, and the following table highlights the key changes from the eSync to the eSync 2.
|Sync Input||Three ports: high-impedance, low-impedance, and isolated||Four ports: three general purpose and one isolated|
|Sync Output||Two output ports||Four output ports|
|Shutter Glass||Occupied sync input and output ports.||VESA Stereo I/O ports integrated specifically for shutter glass applications.|
|Sync Rate||10 to 250 Hz||30 to 360 Hz|
The general purpose sync-in port on the eSync 2 is designed to utilize high-impedance input. However, it is still possible to input low-impedance signals with an appropriate modification to the input port. Please contact us for details.
Technical drawing of eSync
Technical Drawing of eSync 2